Fluoroscopy & Radiography medical 100mA x ray machine for sale MSLCX08:
The unit can used for general radiography,Bucky radiography of body's head,chest,foot and hand,bones in lying-medical x ray machine for sale
2,single-table single tube.
3,single-phase,full wave rectification.
4,stationary anode x-ray tube assembly.
5,X-ray tube filament s voltage regulator,space charge compensator.
6,Radiograph plain table wimple structure and easy assembling.
Single table and single tube
Design according to the electric mechanism of middle-size machine,The function is reasonable
The high-tension system is of split structure.Equipped with afullwave rectification high-volage generator and independent X-ray tube unit,with large output power and reliable and stable operation
Supply voltage (V),roentgenoscopy,radiography (KV) and stepless adjustable
Equipped with the manostat for filament of X-ray tube and the spacecharge compensator as to ensure the stable and accurate output
Equipped with the capacity limitation and portection circuit and KV mA S interlock limitation and protection
Adopt the digital circuit timer,grade according to R10 priority coefficient and can control time exactly,
Diagnostic table is of electric back rotation and is equipped with vibrating filter-ray device
The spot film device of luminescent screen can hold the film jacket of various sizes and can make the photography of full-film and two sectional films separately.
The column post hast has the bottom track only,which is easy while installation.
Demand of power supply:
Voltage,Frequency:380V/220V ±10%50HZ ±0.5Hz
Intemal resistance:2Ω(380V) 0.6 Ω(220V)
Power supply:Adjustable range:380±10% at220V being adjustable continuously
current:Small focus: 15mA,Big focus:30mA 60mA 100mA
time:0.04~6.35s 23grades together grading a according to R10 coefficient
Max DC output voltage:90KV
Max DC output current:100mA
Focus:Small focus 1.8×1.8mm big focus4.3×4.3mm
Length×width×height of bed surface:2000mm×665mm×700mm
Moving range of bed surface:640mm in Length,200mm in breadth
Ray-filter of table:
Travel in length ≥50mm
Grid density N=28
Convergence distance f.=100cm
Grid ratio r=8
Column of X-ray tube unit:
Moving Longitudinally along radiographic table:1000mm or 1800mm
Distance to friend as moving up and down (along column):650~1800mm
Rotary around the center of cross arm:±180°
Rotary around the shaft line of X-ray annular tubes :-10°~60°~ +120°
Max size of cassette:356mm×432mm(14″×17″)