Trolley: Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
Drawer unit ONE drawer totally extractable
Gas requirement: Medical oxygen and nitrous oxide with a pressure ranging from O2: 0.32~0.6MPa ; NO2: 0.32 MPa to 0.6 MPa.
O2 absorb capacity 1.5KG
Flow meter O2:0.1~10L/Min, N2O:0.1~10L/Min
oxygen concentration in the mixed gas N2O/O2: > 25%
Oxygen Flush: 25~75 L/min
Respiration Mode Manual
Pressure limit range: 0 ~ 6.0 kPa
Alarm: O2 pressure is too low
Ambient temperature: 10 ~ 40oC
Relative humidity: no higher than 80 %
Atmospheric pressure: 860 hPa ~ 1060 hPa
Air source requirement: Medical oxygen and laughing gas with a rated pressure ranging from 0.3 to 0.5MPa.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 22.214.171.124 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.